Imperfections of H-coils can cause CW-ACW differences in measured rotational power loss
What is it about?
This paper offers an explanation of the possible mechanism behind the CW-ACW differences. It was shown in reference  that H-coils do not have perfect immunity to off-axis components of magnetic field and detect some erroneous signals caused for example by the normal field (perpendicular to the sample surface). Theoretical analysis shows that such erroneous signals can responsible for the CW-ACW differences.
Why is it important?
The CW-ACW differences are known to be significant at high rotational flux density (above 1.5T for electrical steels). This effect is well documented and explained on the basis of angular misalignment between the B and H sensors, as described in the references [13, 14]. However, there was no explanation of the CW-ACW differences which are still present at lower excitations (1.2 T and below). This paper explains that such effects can arise simply due to the off-axis sensitivity of the H-coils. The power loss averaged from CW and ACW measurements is not affected, because the errors cancel each other.
The following have contributed to this page: Stan Zurek
In partnership with: